For diagnostic use only when a workstation is not available
To ensure the safe and optimal display of medical images, please adhere to the following safety guidelines:
Ensure that you have received the appropriate training to read digital medical images
Ensure that you have reviewed the product's user manual
Do not use this product while under the influence of alcohol or drugs
Reading environment should be free of distractions
Ensure that ambient lighting conditions do not affect the screen display
Ensure that the screen display is free of glare and noise when your line of sight is perpendicular to the screen
Ensure that the wireless signal to the device is strong enough to maintain a good internet connection
Screen should be free of damage, dirt, and dust
Screen protectors should be temporarily removed for the duration of the viewing session
Automatic screen brightness options should be disabled
Screen brightness must be calibrated using the Light Test prior to reading
Do not use this product when the device has entered a low battery state
To ensure the safe and optimal display of medical images on mobile devices, please follow these instructions to calibrate the screen brightness prior to diagnostic reading:
Disable all automatic dimming and tone adjustment options.
Turn the brightness down to its minimum level.
Slowly increase the screen brightness until you can view all low contrast circles inside each square.
Note: If the circle inside the darkest square cannot be visualized, then the screen brightness is too low. If the circle inside the lightest square cannot be visualized, then the screen brightness is too high.
Please enter text you want to place on the image.
Please enter comment on Annotation you want to save.
Please enter comment on Key Object Group you want to save.